Hey Lauren. Hope you are doing well. Good to see you on here.
We have moved all of our DIBH electron boosts to the free breathing scans as the others have indicated. If we feel the boost absolutely needs to happen on the DIBH scan we will setup as usual using the free breathing scan, have them take in a deep breath to the breath hold surface, then insert the cone and snap a new reference under a new boost field. That reference basically tells you what the cameras can and can’t see and you can set your ROI appropriately for monitoring. I can say since we stopped using the DIBH scan for boosts we have probably only “needed” to do this 1-3 times in 3+ years.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.