No, to answer your specific question, the VRT reference capture can not be converted to an RT Structure set.
You can capture a VRT reference capture in treatment mode, if your patient is set up correctly, i.e. verified by imaging, ensure the appropriate region of interest (ROI) is drawn and use the VRT reference surface for set up and monitoring purposes.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.