In reference to which MV Iso option to use, “Field center” option will calculate centroid of your 10×10 field and use that as radiation isocenter. “Image center” option uses the calibration of your MV imager to establish radiation ISO. VRT trainers instructed that it is up to us to decide, but usually “image center” is used if you have ISOCAL, which is apparently something on TrueBeams. Since we do not have ISOCAL, I am planning on using the “field center” option, but will probably use MLC to define my 10×10, as I trust MLC tolerance more than jaw tolerance.
Hope this helps.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.