Yes absolutely as long as you complete a planning scan in FB and then in BH so you can generate both data sets. Using the FB surface for set up is standard practice in most sites.
Physics/dosimetry just need to send both surfaces over labelled FB and BH so you can clearly differentiate between the two. One of VisionRT’s clin apps specialists will be able to talk your team through how to set this up, then you can use FB for set up, switch over in the linac to the BH surface, coach your pt into BH, and adjust any minor lat and long deltas in BH prior to trt.
Hope that helps
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.