(2) Important question with no easy answer. First you must find out if your shift e.g. >1mm at 90/270 is due to patient shift or due to couch run out. I.e. your machine QA should give an answer to the amount of couch runout when rotating from 0° to 90/270. If this value is known, then you can test AlignRT with a head phantom where you know that you have no patient movement. Here the delta should be no larger than the known geometrical couch runout. I did these kind of tests both with a head phantom and 2 volunteer RTTs. Translational deltas where <0.5mm for the full range from 90 to 270 for both phantom and volunteer. rotational deltas where up to 0.4°. So I don’t have any indication for artifacts of AlignRT at extreme couch angles.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.