Over 3 years, approximately 300 patients were treated with over 5000 treatment fractions using this workflow. There were no wrong-site treatment errors. Approximately a dozen near-miss events related to the daily setup process occurred, largely on the first treatment. Root-cause analysis attributed errors to user origin misidentification, couch parameter miscalculation, incorrect immobilization device use, and immobilization device indexed at the wrong indexing position. Skin marks and tattoos were unnecessary. Continuous quality improvement added additional quality assurance checks, resulting in no near-miss incidents or adverse events in the preceding 12 months.